Recently, a customer inquired about the connection of DR digital flat-panel detectors. Today I will explain the connection of the wired DR digital flat-panel detectors.
1. The DR digital flat panel detector power supply and network cable connection: insert the AC power plug into the AC power socket of the adapter, and connect the DC output terminal of the adapter to the power interface (POWER) of the control box.
One end of the network cable is connected to the network interface (LAN) of the control box, and the other end is connected to the network interface of the workstation
2. High voltage interface cable connection of DR digital flat panel detector
Connect the detector end (DB15 plug) of the high-voltage interface cable to the control
Connect the box to the HVG port and tighten the screws.
Connect the high-voltage generator end (crimping pin terminal) of the high-voltage interface cable
Connect according to the port type of the high voltage generator.
Follow the above steps to complete the connection of the dr digital flat panel detector. If you have other needs, please call us.