The vertical bucky stand can be used with a DR Wireless plate detector. Recently, Mr. W from Anhui*Medical Co., Ltd. consulted about the vertical bucky standfor use with a mobile DR. Currently, there is a mobile DR in the hospital. They want to have a vertical chest stand. The flat panel detector is wireless. The size is 17 * 17 inches. Regarding the thickness of the flat panel detector, the customer is unknown. It is recommended that the customer and the hospital determine the thickness, because there are many types of vertical bucky stand. The following is a detailed introduction to the thickness of the card slot, the vertical chest film of the front-mounted film The thickness of the rack slot is 2CM, that is, the thickness of the board is not more than 2CM. The thickness of the vertical chest film rack slot of the side-mounted film is 2.8CM, and the thickness of the board is not more than 2.8CM; the thickness of the vertical breast film rack slot of the DR film box is 3CM, and the thickness of the board does not exceed 3CM.
Among them, the vertical bucky stand for front-mounted films is suitable for cassettes, CR IP panels and wireless flat panel detectors, the vertical bucky stand for loading films and the vertical bucky stands for DR film boxes are suitable for cassettes, CR IP boards and wireless Or wireless flat panel detector. The grid can be installed on the inside of the front panel of the front-mounted vertical chest film and the inside of the door of the side-mounted vertical chest rack, but the DR vertical bucky standcannot be installed with the grid. Seeing this, everyone is more clear about the difference between these three products.
Regarding the fixing method of the mobile base, it can be directly fixed on the ground with expansion screws. We call it a fixed vertical bucky stand. Customized according to the needs of different users. If you still don’t understand, please call us and we will answer them one by one.